Discussion in 'Other Equipment' started by Robert Squire, Dec 14, 2017.
Call me please.
804 five one 3 7153
Broken element or bad connection at top of element. See the "open circuit" light?
Thank you Scott I ll give you call tomorrow morning
If your amps are showing zero, there is an incomplete circuit with your heating elements. Whenever I have this happen, this is my checklist. First thing, check to make sure no heating elements are broken, open top cover and use a multimeter to check for continuity. If all is good, check to make sure all heating element clamps are tightened well. If it still doesn't heat, take off each clamp one by one and check for any corrosion along the tops and clean with wire brush and retighten. Hope this helps!
Sent from my iPhone using Tapatalk
Separate names with a comma.